|Place of Origin:||China|
|Minimum Order Quantity:||1000 PCS|
|Packaging Details:||100 pieces packed in a nest box|
|Supply Ability:||500000 PCS per Month|
|Product:||Drug Test 5 Multi Urine Card||Accuracy:||>99%|
|Product: IVD Medical diagnostic Drugtest 5 Multi urine rapid test dipcard||Instrument classification:Class II|
|Specimen:Urine||Shelf life:24 months|
|Reading time:3-5min||Parameters:5 kinds of drugs|
|Storage:Room Tempreture||Logo:Customer's Logo|
|Certificate:CE,ISO,FDA||Shipping:By sea/By air/TNT/Fedex/DHL|
Multi-Drug Urine Test Panel is a qualitative diagnostic lateral flow urinary assay for the detection of drugs of abuse in human
urine at cut-off levels specified by SAMHSA in a convenient one step multiple test strip format.
1. Total 16 drug substances can be freely composed into multi-drug test panel.
2. Cut-off levels comply with international standards.
3. Read results at 3~5 minutes.
4..Store in room temperature and valid in 24 months.
1.Each pouch contains.
2.One DOA test device.
1.Remove the test card from the sealed pouch and use it as soon as possible.
2.Remove the cap from the end of the test card. Immerse the strip(s) of the test card vertically in the urine specimen with
arrows pointing toward the urine specimen. Immerse the test card to at least the level of the wavy lines on the strip(s), but
not above the arrow(s) on the test card.
3.Remove the test card after 10 seconds and lay the card flat on a non-absorbent surface, and then begin timing.
4.Wait for colored lines to appear. Interpret the test results at 3-5 minutes. Do not read results after 10 minutes.
Interpretation Of Results:
Two lines appear. One red line should be in the control region (C), and another apparent red or pink line adjacent should be in
the test region (T). This negative result indicates that the drug concentration is below the detectable level. (*Note: The shade
of red in the test line region (T) will vary, but it should be considered negative whenever there is even a faint pink line.)
One red line appears in the control region (C). No line appears in the test region (T). This positive result indicates that the
drug concentration is above the detectable level.
Control line fails to appear. Insufficient specimen volume or incorrect procedural techniques are the most likely reasons for
control line failure. Review the procedure and repeat the test using a new test panel.
Newlystar-Medtech is a well established and professional supplier of Finished Pharmaceutical, Active
Pharmaceutical Ingredient(API) and Medical Devices. With a proven track record of passed years,
Newlystar-Medtech has been recognized by her customers as a high-quality and reliable supplier not
only on products, but also on assistance for audit and improvement of GMP level, quality management
system, EHS management system and registration dossier support, etc.
The essence of Newlystar-Medtech's activites, is a commitment to her customers with professional
service and high-quality products with effective cost and innovative solutions.
All pharmacists work in Newlystar-Medtech like a bridge link the customers and Chinese manufacturers,
especially, their professional working is to make the formulation manufacturers who is familiar with
Chinese Pharmacopoeia to understand well the international requirement,to ensure the success of
The advantages on professional working and strongly representative base on first-class manufacturing
partners in China, enable Newlystar-Medtech to provide her customers with a wide range and high-tech
products, best services and capabilities for their actions in tender, distributing and improvement in their
national healthcare system. Newlystar-Medtech also dedicates herself to sourcing quality medical
products for Chinese market.
Why Choose us?
1. Custom Logo Accepted
2. Excellent service and competitive price
3. Sufficient supply